Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
|Published (Last):||9 January 2017|
|PDF File Size:||3.32 Mb|
|ePub File Size:||6.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Exceeding this limit may cause damage to the KP SB.
Plug in the power cable. Connect the digital multimeter to A13J2 pin4. Set the output amplitude of the Signal Generator to 0. Only the Signal Generator, a bus controller, and an HP-IB interface with appropriate cabling are required to perform these procedures.
Set the Siganl Generator as follows: Mamual normal terminators carriage-return, and line-feed are considered further data.
Frequency accuracy and stability are dependent on the reference source being used, either the internal 50 MHz reference oscillator or an external source that operates at I, 5, or 10 MHz.
Resets to 0 Comments Performing this feature does not affect any saved front panel set-ups. Table lists the ratings and the HP part number of the line fuses.
If the Signal Generator is in local operation when it is enabled to remote operation that is, REN is set true and it receives the Local Lockout message, it switches to remote with local lockout operation the first time it is addressed to listen.
Set the function generator as follows: Leave the carrier frequency set to MHz. Indications The carrier frequency changes by manul stored value 856b frequency increment. This begins with a valid final terminator keystroke or data entry and ends with the execution in the hardware.
Example Recall the front-panel set-up saved in memory location It is kanual that the Signal Generator is able to both handshake and recognize its own address.
Hewlett-Packard B Signal Generator
Any attempt to send data to the Signal Generator delays bus operation until the execution is complete. The reference is established on the spectrum analyzer display.
Replacement A4 assemblies are shipped from the factory without the attenuator pad installed. Figure also shows the cable connections for this measurement. Fabrication depends upon machining and assembling to 856b tolerances.
Press the [shift 1 key then four data keys.
When the pulse waveform from the external source goes high, the pulse output from the HP B turns on. Step the frequency down from to MHz and record the deviation at each of the 10 MHz steps. Set digital multimeter Range to V.
Full text of “HP: B B final”
Indications The carrier frequency phase changes in I degr ee steps each time the keystroke sequence is followed. These charges are generated kanual numerous ways such as simple contact, separation of materials, and normal motions of persons working with static sensitive devices. A summary of program codes is provided in Tables and An FM deviation change typically occurs 60 ms after the final terminator Units Key triggers the change.
8656n one conduc- tor of a two conductor outlet is not sufficient protection. Detailed Operating Instructions Dlnstrations in this section of the manual, provide you. The dB and mV annunciators will now he off and the dBm annunciator will be on. Connect mqnual as shown in Figurefor the signal path shown as below. Exceeding this limit may cause damage to the Signal Generator. Whenever it is likely that the protection has been impaired, the Signal Generator must be made inoperative and be secured against any unintended operation.
The detailed operating instructions provide comprehensive information about all of the Signal Generator functions and features. Describes any miscellaneous information or information to clarify the feature or function capabilities. manuall
Each filter is then adjusted for a minimum signal output. Distortion manuual verified by measuring the demodulated output from the measuring receiver with a distortion analyzer.
Use the [ recall! When handling or servicing equipment containing static sensitive devices, adequate precautions must be taken to prevent device damage or destruction. Press the Test Files softkey.
HP 8656B Manuals
Software execution time is required for manuao microprocessor and related digital circuits to process the data. This manual provides the best information we could mnaual. Subsequently pressing this modulation key switches on the modulation and recalls the previous modulation value. Change the Signal Generator frequency to The default display will be If the Signal Generator is being returned to Hewlett- Packard for servicing, attach a tag indicating the type of service required, return address, model number, and maual serial number.
About this Manual We’ve added this manual to the Agilent website in an effort to heip you support your product. The deviation displayed on the spectrum analyzer should decrease in relatively uniform steps. Enter an appropriate graph title for your test. Function Vdc Range mV 2.