View and Download Hioki instruction manual online. DIGITAL M? HiTESTER. Test Equipment pdf manual download. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. Hioki Digital M-Ohm HiTester, Hioki – on Sale at the Test Equipment Depot. Slow: less than seconds (at manual ranging). Sampling rates, Fast: .
|Published (Last):||18 July 2015|
|PDF File Size:||17.37 Mb|
|ePub File Size:||5.30 Mb|
|Price:||Free* [*Free Regsitration Required]|
As a result, when comparing an identical sample under the following conditions, a takt time improvement of ms is estimated. Rear Connect test probes.
Command Reference Explains the received data. Don’t have an account?
Hioki 3154 Manuals
Following indicator appears displaying a number blinking. The output queue is also cleared hiooki the power is turned off and turned on again. Connecting Switched Probe Delay time countdown 1.
Queries if condition saved with designated number is to be saved: Setting Judgment Beep Sound SAVE Saves test conditions: Notes On Use Using the product under such conditions could result in electrocution. When setting at auto range, if the test time is set at a short time period, there is a possibility that test may finish without displaying a measurement value because of a range shift. Be sure to apply protective grounding to external equipment to be connected to avoid risk of damaging insulation.
Test voltage Discharge time Approx. This ensures nioki resistance testing is not carried out without proper contact, as this can result in erroneous assessment. Beeps when test duration timer ends. The number identifies saved test condition setting. Page Queries measurement value and judgment result.
Hio,i mode is set for probe mode setting. Quickly discharge residual voltage The post-test residual voltage discharge time has been reduced significantly compared with legacy models. The ST and ST cannot be operated alone.
When disconnecting from thedisconnect test probe first and signal lead last. Other pins are not in hiooki. TEST lamp Turns on during testing.
Insulation Tester ST – Hioki
Indicates that incorrect operation presents an extreme hazard that could result in serious injury or death to DANGER the user. ,anual Testing test Status Page Executing this command during testing results in execute error. Got it, continue to print. Beeps for PASS judgment. COMParator Sets comparator lower limit value and test mode.
RSC does not function Cables are not connected properly. While charging capacitance, time is required for applied voltage to reach test voltage standard value. Don’t have an account? Also connects with INT.
Response is returned in a line with each responding data separated by a semicolon ; Chapter 6 External Interface When timer indicator is blinking, press both key and key at the same manuzl to display “–” blinking or hold down either key or key to display “–” blinking. Beeps for FAIL judgment. Measurement value display decimal position may differ case by case.
Gallery No results were found. BEEPer Sets beep sound during key manul key press press: Measurement Connect test probes.
Hioki 3154 Instruction Manual
Table of contents Table Of Contents Test voltage is V. Connecting Power Cord After maximum data assigned number is reached, it starts from number 1 again. Locking on LOW does not accept “all test start” Saving And Loading Test Conditions Setting Test Signal Off Timing Example These are simple example of the use of the command.
Queries measurement value and judgment result.